You cannot use multiple Digilent® devices in the same DataAcquisition interface. If you need to use multiple devices, add one device per DataAcquisition and start them sequentially.
Digilent devices limit the minimum and maximum allowable rate of sampling based on channel types:
Analog input only: 0.1 – 1,000,000
Analog output only: 4,096 – 1,000,000
Input and output: 8,192 – 300,000
Data Acquisition Toolbox™ conforms to the Digilent Player Mode for the Arbitrary Waveform Generator.
You cannot use background operations with Digilent devices. You can only perform foreground operations.
You cannot perform synchronous and triggered operations using a Digilent device.
You cannot access the digital input and output capabilities of a Digilent device.